Use of Electrosurgical Equipment
1. The integrity of all electrical components must be
assured; all connecting cords, interactive patient plate,
and electrodes should be in good condition.
2. In order to protect the patient from injury and to assure
the best results, pre-lubricated conductive pads should
be applied to the appropriate portion of the patient's
3. Routine certification, at least annually or pursuant to
manufacturers' recommendations, should be done as a part
of the biomedical inspection of this unit.
4. Laser equipment shall be calibrated and checked per
recommendation of the manufacturer.
Approved By Governing Board